Announcments & Equipment Logs
We will announce once the probe is calibrated for its first use with all relevant parameters optimized. Thank you for your patience. NMR Facility Manager 11/23/15
We will announce once the probe is calibrated for its first use with all relevant parameters optimized. Thank you for your patience.
After attending to the tune issue on the probe that is directly caused by users using fingers to turn the turn rods, push past the stop point of the tune rod, or using the tune tool incorrectly, the av300 BBI probe has been repaired and is returned for use. The NMR Facility cannot stress enough […]
Users: The av300 BBI probe has been misused again and is no longer tuning/matching on the 1H, which precludes proton and any 1H-decoupled-X-observe NMR. Until the NMR Managers can take a look, please do not run on the av300 because using the av300 in its current state risks further damage to the console. Please watch […]
Login issues have been resolved on both the av300 and drx499 spectrometers.
An issue similar to that of AV300 is not allowing users to login with their NetID currently. We are working on this and post an update as soon as the issue is resolved.
The UWNetid is not functional and we are working on this. I will provide an alternative or any useful updates on the situation, shortly. NMR Facility Manager 11/16/15
The new gradient capability is fully functional. Combined with the ATMA feature of the porbehead, users can do everything from the computer desktop, including shimming using TOPSHIM. PROBE TUNING : After inserting sample and setting up the correct dataset, just enter atma on the command line. Wait for the auto tuning/matching to complete. SHIMMING : Topshim […]
I apologize for this unexpected delay in implementing gradient upgrade but we hope the wait will be worth it, once users can let the computer do all the shimming. I am glad to report that the major part of this hardware and software upgrade is done and I hope that by Thursday, the 12th, […]